top of page
This tab allows users for setup.
Users may move teeth as desired.
Layer function allows users to stage the setup as well.
Function to create new setup. User may create different setups at same time.
1. Occ Plane
: Step to adjust the occlusal plane
Use 3 control bars to set the plane position
: User sets target arch of the setup
Arch is equal to line of occlusion
Arch is adjusted by 2 control points to move and rotate
Select type of the Arch
3. Tooth Setup
Movement and rotation of the teeth
The position of a tooth can be adjusted
with tooth control dialogue
Properties: IPR, Extraction, Fix
Adjust the controls displayed when a tooth is selected
: This step is to set the position and
properties of the teeth
The tooth control button is at the bottom of the left tool bar
: Choose a tooth and right click to
select between IPR (Mesial) and IPR (Distal)
User may undo their IPR by right click
and select Reset (Mesial) and Reset (Distal)
: Function that automatically aligns all teeth
based on the occlusal plane and arch
5. Auto Align
: Function that automatically push
away neighboring tooth when moved
6. Auto Push
: Function that automatically prevents
all overlaps with neighboring teeth
7. Auto Contact
8. Extration / Fix
: Function to extract or fix teeth
Select a tooth and right click
to open Extraction / Fix menu
Undo extraction: Untick the Ext under
the tooth information table on the left
9. Occ Check
: Displays contacted area of both Mx and Md with colormap
: Click any tooth to change automatically the camera view to front view of the selected tooth
11. Left Tool bar
Z-axis movement option: Adjust the way to move Z-axis
Rotation option: Adjust the rotational axis
Tooth information list: Displays length of each tooth, amount of IPR and extractions
Perpendicular to the occlusal plane or in the root direction
Rotation based on tip or root
: Undo / Redo any movement and IPR
12. Undo / Redo
Click Undo / Redo button on top left corner
Apply rotation of all teeth based on the arch in the occlusal plane view and activate Auto Push to automatically align the teeth Fine tune with tip/torque, intrusion-extrusion
Tip 1: How to Setup quickly
Tip 2: How to move / rotate teeth
without rotating the screen
Move rotation / mesial-distal / lingual - buccal in occlusal plane view and open Tooth Control menu to control Torque
/ Tip / Intrusion-Extrusion.
: Function to add Layers
Layer: Milestone for movements to reach final Setup. User may create multiple layers
General clear aligner production requires layers in order of Expansion > Alignment > Retraction
Add a new Layer
Select Init and click Add to change current layer 1 to layer 2, and copy of Init will become new layer 1
Select any Layer, and click Add to copy selected Layer
Delete selected Layer
Change Layer Name
On the 3D Setup Tap, select New Setup or Edit Setup to go to the Setup screen
Right-click the layer you want to rename to display the Hidden menu
Choose Rename to change the Layer name
: It is a function that can overlap
teeth before and after treatment.
Use the slider bar to adjust the transparency.
Edit / Delete Setup
Edit / Delete Setup
Function to edit / delete selected Setup.
Displays 3D animation from
initial to final layer of the Setup.
Animation can be saved as a movie
Function to display all teeth movement (Mesial/Distal, Lingual/Buccal, Intrusion/Extrusion, Rotation, Torque, Tip)
Function to save the result of the Setup in images, movement amount and bolton ratio in PDF.
bottom of page